Title Electric cooling from room temperature down to
Title Electric cooling from room temperature down to 200 mK M.Tarasov, L.Kuzmin, and I.Agulo, Chalmers University of Technology, S41296, Gteborg, Sweden V.Mikheev, P. Noonan, and A. Adams Oxford Instruments Superconductivity, Old Station way, Eynsham Witney OX29 4TL, UK Outline Building blocks: Pulse tube rifrigerator He3 sorption cooler SIN electronic refrigerator Experimental results Estimated margins, other experiments General view of the equipment Cryostat D300, H850
View of the He3 sorb View of the internal part of the cryostat Oxford Instruments with He3 sorption cooler, removed outer vacuum can (OVC), and removed radiation shields Pulse tube cooler & compressor Sumitomo Cryocooler SRP-052 Two stage pulse tube cryorefrigerator Cold head unit RP-052A1 First stage 20 W at 45 K, second stage 0.5W at 4K Compressor unit CSW-71D, water cooled 7 l/min, W450, L500, H687 mm, 120 kg, Electrical requirement 3 phase 9kVA
Operation pressure 25 bar, steady-state 17 bar Top of the cryostat To reduce interference and noise from grounding we placed our roomtemperature battery feed electronics at the top of the cryostat close to the connectors. Electron cooler chip layout 4 junction structure for cooling/heating at the top and botton Log-periodic antenna for 0.1-2 THz range Double-dipole antenna
for 600 GHz Double-dipole antenna for 300 GHz SPM view of SIN electron cooler SIN cooler with Au trap SINIS cooler Electron cooler with trap IV curves of SIN thermometer 200 Sensor voltage, V C V 100
0 -100 -200 -0.4 -0.2 0.0 Sensor current, nA 0.2 0.4 A 7 k SINIS thermometer IV curve at
Tph=290 mK without cooling (X-es), and under electron cooling (circles) Cooling curves 270 260 B Electron cooling starting from phonon temperatures in the range of 287-365 mK
Electron temperature, K 250 240 230 220 210 200 190 280 300 320 340 Phonon temperature, K 360
380 Ideal SIN tunnel junction IV curve The IV curve of SIN junction have simple analythic form k 1.76Tc eV I (V , T ) 11TTc exp sinh eR T kT The electron power
1.76Tabsorbed k temperature under eV c I (V , T ) 11TTc exp 1 / 5 sinh P eR T kT 5 Te T ph V
Zero-bias resistance with leakage current 1.76Tc T e R ( 0) e R r n Rn Rs 0.6 Te T c
1 Calculated ZBR 1 10 4 . 1 10 rds rds rds 3 n 13 n 25
100 n 50 10 . 1 0 0.1 0.2 0.3 T n .
0.4 Resistance ratio calculated for bias voltages 0, 200, 300 V, thermometer normal resistance 10 k and leakage resistance 35 M Dynamic resistance of SIN 50000 MF355 J9/13 measured 03.03.2003 V0T20 V400T20 V400T250 V150T250
Theory Resistance, k 40000 Dependencies of sensor resistance measured in dilution refrigerator at 30000 20000 20 mK & 250 mK and co ler bias 0, 150, 400 V 10000
Optimal cooling MF355 J9/13 measured 03.03.2003 Electron temperature,K 280 Electron temperature estimated from dynamic resistance at 300V (boxes) 240 200 160
120 80 0 100 200 300 Cooler bias voltage, V 400 500 Calculation of cooling power Cooling power
Pcool 2kb eV V exp 2eRN e kb V2 Pbgn Pcool 5 Effective electron temperature T 5 Rs
Heat balance 4 10 Curve T250 (circles) corresponds to the electron-phonon power transfer 13 Q 98 n Q 96 n 2 10 Q 93 n
Pep=(Tph5-Te5)v 13 at 250 mK, other curves from the left to the right present cooling and heating power balance Q 89 n Q 85 n T250 n 0 Pv=Pcool-V2/Rs-Pbg
0 0.05 0.1 0.15 T n . 0.2 0.25 at bias 392, 384, 372, 356, 340 V Discussion
Obtained cooling down to 190 mK has a reserve of improvement down to 100 mK as in He4 liquid precooled He3 sorption cooler We are still suffering from electric noise coming from high power supply line and its grounding directly to the cryostat via high pressure supply from connector Acustic vibrations also affect operation of electron cooler and cold electron bolometer Operation wth a pulse tube refrigerator at 3.5 K instead of 2.8 significantly prolongs the precooling period and available lowest temperature is 290 mK instead of 275 mK. Conclusion We have demonstrated the first cryogen-free electric cooling from room temperature down to electron temperature below 200 mK. The key idea behind this device is to develop a millikelvin range cryocooler as simple in operation as a conventional kitchen refrigerator. It does not need filling with any cryogen liquid and you need just to switch it on in the evening to have electron temperature of the sample below 200 mK in the morning. The first building block of the device is a double-stage pulse tube cryocooler Sumitomo that provides cooling down to ~3 K. The
second building block is He3 sorption cooler of original design by Oxford Instruments that brings for moderate thermal load a basic temperature of about 280 mK. The third building block is a Superconductor-Insulator-Normal metal (SIN) electron cooler. This type of cooler is analogous to the Peltier effect and in general can provide electron cooling by up to 200 mK. In our very first tests of the whole system we already achieved cooling down to 191 mK. Dream cooler for astronomers A ground-based telescopes Testa Grigia and Heinrich Hertz ...for atmospheric research, ecology, missile detection,... Stratospheric observatory SOFIA
Všude přítomné připravené prostředí Otevřený a vzájemně propojený prostor Cca 20 žáků ve třídě (1-3, 4-6,7-9) Každá třída má svou kmenovou učebnu, dále jsou k dispozici odborné učebny, chodba a relaxační kouty v učebně jsou různá pracovní místa dřevěné police...
Times New Roman Default Design Audio processing using Matlab Sampling A/D (analog in/digital out) Spectrogram D/A (digital in/analog out) Aliasing Filters Example Filter Types Example Filter frequency response Filter order
Lure of Hydropower Big Dams Vs Rebar Presently world's largest Hydroelectric plant On Parana river between Brazil and Paraguay Capacity: 12,600 MW (drought a problem recently): Provides: 25% of Brazil power 78% of Paraguay power Under construction Three Gorges Dam...
Develop a foundational knowledge of PRICING to understand its role in marketing. 4.07. IDENTIFY FACTORS AFFECTING PRICING and PRICING ISSUESOF SEM PRODUCTS. FACTORS AFFECTING PRICING OF SEM PRODUCTS. LEAD TIME. MARKET DEMAND. How much of a product customers will buy...
The analytical framework: Fitness landscapes due to Sewall Wright (1932). Multiple local "hills" (optima) recognized by Wright, studied later by Kauffman (1993), with "complexity catastrophe" arising as more dimensions arise. Applied to institutional evolution by Mueller (2014).
The Afghan Alumni As Shay and Schweitzer (2000) have described, today, the 'Afghan Alumni' operate in four capacities: 1. As leaders of the radical Islamic organizations in their countries of origin (Egypt, the Maghrib countries, Jordan, Saudi Arabia, etc.) 2....
Demonstrate how to resolve nationally at least one chronic global water problem. Can GWF pick one? Resolve at least one long-lasting national water problem within the time frame of the Decade. Is there a role for GWF? Go beyond academic...
Founded in 1986, the Ohio Child Welfare Training Program is a comprehensive, competency-based inservice training (CCBIT) system for staff, managers, and resource families in Ohio's 88 county Public Children Services Agencies (PCSAs). ... SACWIS) Vending machines are located in coat...
Ready to download the document? Go ahead and hit continue!